When you want to start to do something you have never done before, instructions are vital, especially if you are a healthcare worker, a dental professional. And even though a Zoom chairside whitening treatment seems to be incomparably easier than a complicated dental surgery, it requires following the protocol and taking all the nuances into account if you want to receive a painful and effective procedure.
Instructions for Zoom Chairside Whitening System.
Zoom Chairside Whitening Procedure Kit to be used with: Zoom Advanced Power (Zoom 2), Zoom Advanced Power Plus (Zoom 3) and Philips Zoom WhiteSpeed (Zoom 4) Lamps
IMPORTANT NOTE: The Discus Dental Zoom Chairside Whitening System is unique in both technique and formulation. Please read and understand information below prior to initiating whitening treatment.
Prior to use, Zoom Whitening Gel must be removed from refrigeration and warmed to room temperature.
To allow gel to return to room temperature either:
1. Remove syringe from refrigeration at least 6 hours before use (preferably the night before) or
2. If removed just prior to use, place syringe in cup of hot tap water, 120°F (49°C) for 10 minutes. Wait 5 minutes before extruding gel.
1. DEVICE DESCRIPTION
The Zoom Chairside Whitening System is a light-activated tooth whitening treatment for use in whitening discolored teeth.
2. HOW SUPPLIED
The product is designed for one-time usage. It may include Patient Post Care & Maintenance, to be sent home with patient after procedure, containing a tray case to hold custom fitted trays made from EVA material provided, a syringe of Relief® ACP Oral Care Gel for sensitivity relief and a syringe of NiteWhite® ACP, DayWhite® ACP or NiteWhite® ACP Turbo. Refer to Zoom Whitening Gel box for concentration of Hydrogen Peroxide (HP).
3. INDICATIONS FOR USE
Product is intended only for in-office use by trained dental professional for use in whitening discolored natural teeth.
3.1 Candidate Qualification
A consultation is recommended to ensure no underlying oral health issues. You may want to treat existing issues before recommending whitening treatment and disqualify any patient who is perio-involved, exhibits failing restorations or is otherwise in an unhealthy oral state.
3.2 Patients Expectations
Set realistic expectations about potential results based on oral exam. Explain results vary. Most will experience 6-8 shade change on value ordered VITA® Classical Shade Guide.™* Anatomy, calcifications, age, teeth condition, hygiene etc. determine the results each patient will attain. Teeth with yellow and brown stains will achieve greatest success while those with gray shading from tetracycline or other medications may experience less dramatic results but should improve. Crowns and other tooth-colored restorations will not whiten; to maintain uniform color, tooth-colored restorations may require replacement following whitening procedure.
4.1 Use in Special Populations
Research has not evaluated possible effects of whitening procedures on all patient types, Discus Dental recommends the following whitening candidates consult a medical doctor before use:
5. SAFETY DIRECTIONS
Improper isolation may result in burning of gingiva or swollen lips due to UV light coming in contact with tissue or chemical burn due to whitening gel coming in contact with tissue. Patients more susceptible to sensitivity are those with:
7. FIRST AID INSTRUCTIONS
Contact a Poison Information Center or medical doctor at once if large quantity (over 25% of syringe) is swallowed. Provide ingredient label if possible. Do not induce vomiting and immediately give glass of water. If Hydrogen Peroxide (HP) or Carbamide Peroxide (CP) gel gets into eyes, hold eyes open, flush continuously with water for 15 minutes. If contact occurs between whitening gel and soft tissue, apply Vitamin E Oil provided to the affected area. If improper isolation leaves patient’s soft tissue exposed to light, administer a Nonsteroidal Anti-inflammatory (NSAIDs) or Acetaminophen drugs (adhering to the respective Instructions For Use).
8. CHAIRSIDE WHITENING TREATMENT
8.1 (Step 1) Pre-Procedure
8.1.1 Take Shade
Prior to beginning, shade match teeth with value-ordered VITA® Classical Shade Guide.™* B1 A1 B2 D2 A2 C1 C2 D4 A3 D3 B3 A3.5 B4 C3 A4 C4
8.1.2 Diagnose Patient
Diagnose based on dental history, oral health and sensitivity issues (See 3.1 Candidate Qualifications). Set patient expectations accordingly (See 3.2 Patient Expectations).
8.1.3 Treat for Sensitivity
It is recommended to prescribe the following prior to the procedure:
8.2 (Step 2) Preparation
8.2.1 Tray Fabrication
8.2.2 Clean Teeth
Pumice just prior to whitening.
8.2.3 Light Guide
Remove Light Guide and place on lamp. Turn on lamp using power button located on the side of power pack. When the Light Guide is attached, display will read “Guide Status–Guide Attached,” followed by “4 of 4 Sessions.”
8.2.4 Surgical Suction
Place surgical suction tip on high vacuum suction.
8.2.5 Vitamin E Oil
Apply Vitamin E Oil to lips to moisturize before retraction.
8.2.6 IsoPrep Retractor
Retractor provided is necessary to ensure proper positioning of lamp head and additional
protection from light. Insert retractor at an angle, (Fig 8.2a) retracting one side at a time using a dental mirror to assist in placement ( Fig 8.2b). The retractor provided is a universal size and will fit the majority of patients. Small retractors can be purchased separately.
8.2.7 “Before” Photo
Take retracted “before” photograph using a matched shade tab (Fig 8.2c).
8.2.8 Protective Eyewear
Fully recline patient and place Protective Eyewear (Fig 8.2d).
8.3 (Step 3) Isolation
8.3.1 Cotton Rolls
Place cotton rolls in center of upper and lower vestibules (Fig 8.3a). If cotton rolls are too large, unfold gauze and twist like a cotton roll.
Do not substitute gauze. Gauze provided has been tested and proven to provide superior protection. Fully open gauze squares and fold into triangle. Place apex of triangle into posterior cheek (Fig. 8.3b & 8.3c), tuck balance of material into cheek, tucking ends between cotton rolls and retractor (Fig 8.3d).
8.3.3 Face Bib
Carefully place one face bib around retractor, one side at a time. Add additional bib for heavy salivatorius (Fig 8.3e).
8.3.4 Liquidam Dental Dam
Exposed dentin and tissue should be covered by barrier material. Use rule “no pink”. Remove Liquidam syringe, twist to remove cap and attach metal tip. Dry soft tissue and begin by scalloping barrier material to CEJ area (gingival margins) on upper teeth slightly overlapping enamel and interproximal spaces to form enamel seal (Fig 8.3f ). Cure application of Liquidam using curing light, e.g. FLASHlite Magna™ (Fig 8.3g). Liquidam is a resin blend that can become warm upon curing. Therefore, use sweeping motion, moving light back and forth across arch for about 10 seconds. Change to plastic tip. Fill in from cotton roll to just formed line of Liquidam and cure. Be sure to cover all interproximal areas leaving no soft tissue exposed (Fig 8.3h). On lower arch apply Liquidam, using same method. Use tip of syringe to press lightly on application to ensure it is completely cured. Material should be solid and have no give. If necessary, cure arch again for additional 5 seconds. Application should extend distal at least one tooth beyond area receiving whitening gel and be about 2mm thick.
8.4 (Step 4) Procedure
8.4.1 Zoom Whitening Gel
Remove cap of room temperature Zoom Whitening Gel syringe and attach mixing tip by aligning markings and secure with 1/4 turn clockwise. Apply gel to teeth (1-2 mm thick), using blue brush provided (Fig 8.4a). Use caution not to disturb barrier material. Gingival irritation can occur with prolonged exposure to HP gel.
8.4.2 Align Light Guide to Retractor
Once whitening gel application is complete, raise back of chair so patient is in slightly reclined position (~45 degree angle) and able to swallow comfortably while maintaining light pressure on bite block. The Light Guide is designed to ensure proper distance and alignment to maintain correct light exposure on smile zone. Slide lamp into place next to patient. Arm of lamp should be movable, but stable. Position lamp head by aligning slots on Light Guide with retractor (Fig. 8.4b)
8.4.3 Whitening Sessions
Follow prompts on display panel to turn on light and activate timer. Once light is on, a
countdown will be displayed. It is normal for light to flicker for first few seconds. The lamp
will beep once when 3 minutes are remaining and again 3 times on the final 3 seconds of
cycle. When time reaches zero, light will turn off and a long beep will sound. If cycle must
be interrupted, press “Pause” button. Remaining time will flash on display. To reactivate
timer, press only “Pause” button. If step is not done, fourth 15 minute session will be LOST.
Point out to patient that progress of each session will be tracked by 4 green lights on top
of lamp head, each indicating one quarter of a session.
During each 15 minute session, make sure patient has means of communication, e.g. a bell, or do not leave the patient unattended. After each session, gently pull lamp away from patient taking care to not dislodge isolation materials. Carefully remove gel with surgical suction tip and/or wipe surface of enamel clean with damp gauze. Do not irrigate as force of water may pull dam loose and saturate isolation materials. Check isolation materials to ensure “no pink” tissue is visible. Replace or add isolation if needed. Follow same instructions as above for remaining sessions.
Note: For sensitive patients pull back lamp head within Light Guide or attach Light Guide Extender to further increase distance from light. However, please note these activities may reduce results, therefore a fourth 15 minute session may be needed
Tip: If patient notes irritation at margin use Vitamin E Oil to sooth tissue. If patient notes
tooth sensitivity, place Relief ACP found in Patient Post Care & Maintenance, on lingual
surface(s) for remaining sessions.
After last cycle is complete, detach Light Guide from retractor and pull lamp away from patient. Suction or wipe the whitening gel from teeth. Moisten gauze and cotton rolls, remove isolation materials. If barrier material remains interproximally, remove with floss.
9. POST TREATMENT
9.1 “After” Photo
Reinsert retractor, take retracted “after” photo with lighter shade tab in place. Measure shade change by counting shade shift according to value-ordered shade guide (Fig. 9.1a).
9.2 Relief® ACP ORAL CARE GEL
Relief ACP has been shown to decrease sensitivity and remineralize teeth. If patients experience sensitivity, place into take-home trays and seat on patient. Normal wear time is 10-30 minutes. If tray is not available, brush product on teeth. Do not spit for approximately 3 minutes. For best results, do not drink or eat for 30 minutes after application.
9.3 Patient Post Care & Maintenance
Instruct patient on use including Relief ACP and take-home whitening maintenance product.
9.3.1 Take-Home Whitening Maintenance
Use after 24 hours to maintain or to further enhance Zoom Chairside Whitening results. Depending on region, Zoom Kits are available with the following options:
9.3.2 Maximizing Results
10. STORAGE INSTRUCTION
For Gel: Refrigerate gel at 2°C – 10°C (36°F and 50°F). Storage temperature above 10°C
(50°F) will result in reduced shelf life.
Usage Temperature: >60°F/15°C
Has no parallel in the world. Chip is manufactured in our own production in Kyiv, Ukraine.
All the chips we ever sold have been lasting until now.
Initially chip was invented for Zoom Whitespeed by Philips but it also is compatible with Zoom Advanced Power and Advanced Power Plus.
Установить чип Bleach Infiniter может любой специалист клиники. Требуется только отвертка.
Среднее время установки чипа в лампу - 40 секунд.
Ваша лампа остается на гарантии завода изготовителя Philips.
We have been using Philips equipment for teeth whitening for 5 years, but the guides seemed very expensive, because we often need to buy new Whitespeed Chairside Kit. We bought Infiniter which made guides for zoom bleaching much easier and cheaper. Such decision allowed us to buy only gels for teeth whitening, and we do not pay for the chip. I want to thank Bleach Infiniter for such innovation, for such an adapter, and for the fast delivery.
As any businessman wants to save, but at the same time that would not be damaged quality. In this we helped Bleach Infiniter for Zoom, which can unlock WhiteSpeed and give an unlimited number of cycles. Thanks to that guides for zoom whitening, we saved a large amount of money and did not lose in quality. It works on all modes (Low, Medium, High). At work there were no failures and censures, we get a chic result. For a year the battery-powered version was working, and for the New Year we received a new version as a gift - guide for zoom dental (chip without batteries). It is more convenient. The acquisition of Bleach Infiniter is one of the most profitable investments lately.
My name is Arik. I'm a dental doctor therapist in Moscow. We have been using Zoom lamp for about 2 years. Four months ago, we bought a special adapter. What is the purpose of the adapter? It gives us the ability to use the Zoom lamp without any restrictions and guides for zoom discus. It means we don't have to buy a full bundle (gel and chip) for each procedure anymore. We purchase only gel. So, we have successfully reduced the cost of the bleaсhing procedure. I will demonstrate you how it works. At the moment, the zoom is turned off. We turn zoom lamp on to find out the lamp is not working. Now, we activate the adapter, and we get a working lamp! That is how we reduce the costs of bleaching procedure. It will make the process of teeth whitening more available for customers. We are thankful to Bleach Infiniter for giving us such opportunity. Good luck!
I was advised to take apart Philips Zoom and attach the lamp switcher, but I did not want the technician to adjust such a complicated and expensive device. The colleague who has already tried Bleach Infiniter advised me it by proving there is an external solution without the need to get into the details. I am already using Infiniter for more than a year, and I am satisfied with its functionality like original guide for zoom discus dental.
Isn’t it wonderful that there is a device capable of solving the problem of limitation? It is a very favorable offer in terms of convenience and price, especially since the guys offer a discount for technicians. I’ve already installed Bleach Infiniter to some of my regular customers and help them with expensive guides for zoom advanced power. They are pleased with its work, which allows me to gain some extra money.
Answer: Chip was initially invented as a reliable and long-lasting module. We had made multiple number of tests to make sure that a chip lasts in non-stop mode 24/7 for many years. Warranty for Bleach Infiniter is 10 years but we are sure that chip will last much longer.
Answer: Yes, Your Zoom light will become totally unlimited until the unlimited chip attached. You can easily detach a light guide with Bleach Infiniter installed and upgrade your Zoom light to a newer model from your supplier. A chip does not affect on your device, it simply gives a response to your device about how many cycles still available.
Answer: A chip does not affect on your light in any way since it does not require any consumption, it’s an external solution. Just like a Sim card on the phone, it only has a software which gives a response about how many cycles still available.
Answer: Yes. After you detach an unlimited guide, your device becomes limited as before. (There is no way to check that unlimited chip ever been used with your light.)
Answer: We offer PayPal payments system to guarantee the fulfillment of obligations by the seller. In case the customers do not accept any conditions or stay dissatisfied, they can request their money back.
In the case of payment of purchases to be delivered, the money will not be credited to the seller’s account until the buyer confirms the delivery, or the dispute’s opening time expires. Within 180 days from the moment of payment, the buyer has the opportunity to open a dispute if the delivered goods differ from the ordered one, or the delivery has not been carried out. If the buyer and the seller could not come to a mutual agreement, the dispute can be transferred to the claim. A claim may be placed within 20 calendar days from the date the dispute was opened. If PayPal decides in favor of the buyer, the full purchase price, including shipping costs, will be refunded.
Read more about user protection and security on the official PayPal website: www.paypal.com/security
Answer: Yes, Infiniter works with all versions including:Zoom 2.Zoom 3 Advanced Power.The latest version Zoom 4 WhiteSpeed with LED lamp.
120 minutes long tests and demonstration video of the chip on all intensity modes. We show multiple numbers of short and full 15 cycles. The device shows 4 of 4 cycles available after each cycle.
To confirm the continuity of filming and the authenticity of the video, there is a clock next to the lamp.
You are free to choose the whitening gels and consumable trademarks: any of them can be used.
Most dentists, who have been already working with an unlimited chip, continue to use the original Philips gel (whitening kit). With this approach, they use the gel from the kit on two patients for 4-6 procedures and thus make the bleaching procedure itself cheaper.
But there are also analogues of the gel, which are cheaper and are successfully used in practice.
We would like to draw your attention to those alternative bleaching material which is widely used by our current customers and already proved it’s worth in daily Zoom whitening performance.
Beyond MAX whitening kit;
Belle MAX Bleach Bleaching System;
Amazing White Universal whitening kit;
WSD Labs USA;
As you can see, ZOOM gel can be purchased at a lower price than Philips kits. At the same time, the cost of the whitening procedure will cost your dental practice several times cheaper. So when using the Amazing White Universal gel, the cost price of one bleaching procedure will be $15 only
It’s your decision which light-activated bleaching material to use but we know for sure which chip you need - the chip Bleach Infiniter! Order now and start saving!
The prime cost of the original 1 patient Zoom kit achieves $120. Without our chip, the clinic is forced to spend this particular amount of money each time it buys a new kit
After installing Bleach Infiniter for zoom, you can buy any other whitening paste designed for a larger patients’ turnover. It will be much cheaper than Zoom Chairside Kit - just calculate the price per procedure! For instance, Beyond MAX 5 is designed for 5 patients, and its price is up to $115, which means the prime cost of one session per patient is up to $23.
An intern has accidentally chosen to start the procedure of teeth whitening on Philips Zoom, but the patient was not served. At the same time, the procedure was written off. The trainee was reprimanded, and a month later, another girl repeated the same mistake.
The doctor was upset and complained to the engineer. He wondered how it was possible to make that sort of mistake. He could not get why such devices do not have any protection against the accidental operations.
A computer engineer wanted to decipher the operation protocol just for fun. As a result, after 2.5 months of restless night shifts in 2015 he managed to develop a device adapter that solved the problem of consumables.
Local hospital management shared this innovation with its friends from the other clinics. The news about this adapter spread very quickly. Soon, other hospitals asked the engineer to make the same adapters for them.
During 2016 and 2017 years we were trying to minimize Infiniter that has size 17*37*63 mm and it has wire and 3 AAA batteries.
In September 2017 we developed new product Bleach Infiniter Ultra. Bleach Infiniter Ultra is only small electronic board/chip with size 1*8*18 mm.